The How To section in the Help menu of the app provides several resources to support you- for instance short animations or detailed written instructions. Please also check our supporting materials online: www.sigmaaldrich.com/mquant-stripscan
If there is an issue during the camera acquisition of your test strip, a message at the bottom of the screen will usually explain what to do. For a successful readout, please check the messages and follow its instructions.
Here are some general tips to obtain robust results:
The phone camera: Make sure your camera functions properly: Check if the camera lens is dirty, if your camera shows signs of focus issues, that your image section is the correct size to allow alignment of the crop marks, and that your crop marks on the viewfinder of your phone screen are aligned with those on the reference card.
Reference card: Make sure your reference card is in good condition: It should not be scratched, bent, or stained. Photocopied cards will not work. Also check that you are using the correct Reference Card for the parameter that you want to measure.
If you find that your card is damaged, you can acquire replacements at www.sigmaaldrich.com/mquant-stripscan.
Lighting: Make sure to provide good lighting conditions: Are you measuring in an area with little or very bright light? Are there reflections or shadows on the card? Diffuse lighting and medium brightness are ideal for measurements.
Strip placement: Make sure you place the strip with the reaction zone facing upwards and aligned with the marks of the Reference Card.
We are just getting started with StripScan. You can look forward to receiving more functionality as we continue to expand the app for the MQuant® portfolio- We will keep you posted for updates!
You can buy MQuant® StripScan Reference Cards online from late summer 2018 (see e-shop links on www.sigmaaldrich.com/mquant-stripscan). Please also check your local retailer.
No, the MQuant® StripScan system was specifically calibrated to work with the color reactions obtained by MQuant® test strips (and pH strips formerly marketed as MColorpHast®, now MQuant®). While the use of other manufacturers’ strips may yield result values, these values will not be correct since they have not been calibrated with the app algorithm.
No, the MQuant® StripScan system was specifically calibrated to work with the color reactions obtained by MQuant® test strips. While the use of Reflectoquant® test strips may yield result values, however these will not be correct since they have not been calibrated with the app algorithm.
The color of the test strip may intensify over time, and the readout works in a defined time frame. A progress bar at the bottom of the acquisition screen lets you know how much time is left to perform the readout. It is important to use a fresh strip when repeating the measurement.
Please make sure the Reference Card corresponds with the test parameter you selected in the app. To obtain Reference Cards, please see the e-shop links on www.sigmaaldrich.com/mquant-stripscan, or check your local retailer.
MQuant® test strips work within a certain concentration range, as stated on the label of the test strips. If the analyte concentration in your sample exceeded this range, you can dilute or concentrate your sample, or choose an alternative MQuant® product with suitable range.
Please make sure your phone’s GPS is turned on, and that StripScan is allowed access to GPS data.
We are working on it! As of now, StripScan is available in English only.
The pH is given in gradations of 0,5 pH units.
The nitrate values are displayed in the following gradation:
0 – 5- 10 – 15 – 20 – 25 – 35 – 50 – 75 – 100 – 250 – 500 – >500 mg/l.
Make sure you have internet connection, and try synchronizing again by pulling down on the result list.
If you have already finished incubation of your test strip, or for serial measurements, you can override the reaction countdown. If you skip the timer however, make sure to monitor the correct reaction time yourself.
MQuant® StripScan is currently available for iPhones with iOS 10 and higher.
In the StripScan app, you can discard a result only immediately after acquisition. To manage, share and delete your results, please use StripScan web (registration required).
Strips from production year 2018 and later will work with MQuant® StripScan. Older batches may also function with MQuant® StripScan; however, please compare the results with those of a current batch if you want to use the app with an older test strip batch.
Make sure your phone is connected to the internet. You can synchronize the data manually by pulling down on the result list in the app.